Proper breathing is the single easiest way to help your patients improve treatment with Ventavis® (iloprost) Inhalation Solution.
The patient should sit in a comfortable, upright position and pick up the I-neb Adaptive Aerosol Delivery (AAD) System with the display screen facing down. The mouthpiece should be held at a 90-degree angle to the mouth, like holding a hamburger; your patients may rest their elbows on a table while holding their I-neb AAD System. The I-neb AAD System should be held in this position for the entire treatment session.
Counsel your patient to keep the lips closed around the I-neb AAD mouthpiece when breathing in and out.
Because the drug is delivered only during the first 80% of inspiration, longer breaths mean that more Ventavis will be delivered, decreasing treatment times.
|Patient Breathing Technique|
|Step 1||Sit down comfortably in an upright position.|
|Step 2||Turn the I-neb AAD System on.|
|Step 3||Hold the I-neb horizontally with the screen facing down.|
|Step 4||Seal your lips around the mouthpiece.|
|Step 5||Breathe in and out through your mouth in a slow and relaxed manner.|
|Step 6||You will feel a vibration through the I-neb each time medication is delivered.|
The I-neb AAD System will sense the patient's breathing pattern on the first 3 breaths and begin to deliver Ventavis on the fourth inhalation. When Ventavis delivery begins, the I-neb AAD System will vibrate each time there is an inhalation. This is normal and indicates that Ventavis is being delivered.
If, during treatment, the I-neb AAD System sounds 4 short beeps, it means that the I-neb AAD System is being held improperly, either at too high or too low of an angle. At the same time, the I-neb AAD System will display error code 66. The patient should reposition the I-neb AAD System at a 90-degree angle. This will reset the device and remove the error code. Treatment should then be continued.
Positioning oneself in front of a mirror to get a visual image of what a 90-degree angle looks and feels like can be useful when first starting treatment. At the end of each treatment, the I-neb AAD System buzzes to let the patient know that the treatment session is complete and a smiling face will appear on the I-neb AAD System screen.
If your patient needs a break, then she can take one. Have your patient rest a minute or two then restart treatment. Remember, it will take the I-neb AAD 3 breaths in and out to adapt to your breathing pattern before Ventavis will be delivered again. As long as treatment is started again within 10 minutes, the treatment will commence from where the patient paused.
|Tips for breathing success|
Take LONG and SLOW breaths.
Remind your patient to keep the lips around the mouthpiece when breathing in and out.
|III||Have your patients relax and not work too hard. SLOW and STEADY works best.|
If your patient needs a break, he/she can stop, take a rest, and then start again.